Pure water glass electrode with flexible measuring range, stable reference electrode, long electrode life, easy to install, high quality low noise cable enables long distance signal transmission.
Resistant to strong acid and alkali corrosion, online pH measurement
- Adopt ceramic sand core liquid junction, two-liquid junction electrode, effectively extends electrode service life.
- The shell material is special glass, high temperature resistant, acid and alkali resistant.
- The gel electrolyte does not require additional electrolytes.
- Accessories: PTFE sheath or stainless steel sheath. Increase the acid and alkali resistance of the electrode, easy to install.
- High-quality low-noise cable can achieve 60m long-distance signal transmission.
pH Electrode is widely used in the continuous testing and monitoring of pH, ORP and temperature in thermal power, chemical fertilizer, metallurgy, environmental protection, pharmaceutical, biochemical, food, drinking water, etc.
Q1: What is a pH electrode?
A1: The pH electrode is also called the pH probe pH sensor. The pH electrode is the part of the pH meter that is in contact with the substance to be measured and is used to measure the electrode potential.
Q2: Why do pH electrodes need to be soaked?
A2: The pH electrode must be soaked before use, because the pH bulb is a special glass membrane with a thin hydrated gel layer on the surface of the glass
membrane, which can only interact with the H+ ions in the solution under fully wet conditions. the response to. At the same time, after soaking the glass
electrode, the asymmetric potential can be greatly reduced and tend to be stable.
Q3: What is a reference electrode?
A3: An electrode with a known and constant electrode potential that does not respond to the activity of hydrogen ions in solution is called the reference electrode. Reference electrodes include mercurous sulfate electrodes, calomel electrodes, and silver/silver chloride electrodes. The most commonly used are calomel electrodes and silver/silver chloride electrodes. The role of the reference electrode in the measurement cell is to provide and maintain a fixed reference potential, so the requirements for the reference electrode are that the potential is stable and reproducible, the temperature coefficient is small, and the polarization potential is small when current flows.
Tips: What is a pH combination electrode?
An electrode that combines a pH glass electrode and a reference electrode is called a pH composite electrode. If the outer shell is plastic, it is called a plastic shell pH composite electrode. The outer shell is glass, it is called a glass pH composite electrode. The biggest advantage of composite electrodes is that they are combined into one and are easy to use. The structure of a pH composite electrode is mainly composed of an electrode bulb, glass support rod, internal reference electrode, internal reference solution, shell, external reference electrode, external reference solution, liquid junction, electrode cap, electrode lead, socket, and so on.
Thank you for buying industrial test and measurement equipment on sisco.com, all products sold by sisco and the partner cover a 12 months warranty, effective from the date of receiving the products.
What is covered?
sisco is responsible for providing free spare parts, and free technical support to assist the customer to repair the defective products until the problem is solved.
What is not covered?
- Product purchased from anyone other than a sisco store or a sisco authorized reseller.
- Expendable parts.
- Routine cleaning or normal cosmetic and mechanical wear.
- Damage from misuse, abuse or neglect.
- Damage from use of parts other than sisco approved.
- Damage from use outside the product’s usage or storage parameters.
- Damage from use of parts not sold by sisco.
- Damage from modification or incorporation into other products.
- Damage from repair or replacement of warranted parts by a service provider other than a sisco authorized service provider.
- Damage caused by the application environment not meeting the product usage requirements and the failure to perform preventive maintenance.